The DF-1P megapulse Tester is designed to generate the waveform described in AAMI DF-1, Figure B.2., for the connector cavity current-carrying test in AAMI DF-1, Annex B. The test can be set to automatically conduct the first 490 cycles (500 ± 50 as show in standar)of the test, and then stop testing to allow laboratory personnel monitor the last 10 cycles in accordance with the Standard. Voltage and current BNC outputs make waveform analysis convenient.
A front-panel switch reverses the output polarity. Additionally provided with BNC oscilloscope output for voltage (100:1) and current (1V/A) for output monitoring.
The waveform output is 1.05uSec rise time, with a 18mS duration, with an output impedance of 15 ohms.
MegaPulse DF-1P Current carrying Tester ships complete, ready to perform the DF-1 test:
Create an Account
Already have an account? Login